CHROMA 6314 USER MANUAL PDF
The Chroma A W and A W electronic load mainframes accept the user-installable A series load modules for easy system configuration. user can perform on line voltage measurement controlled instructions via RS- C or GPIB The Chroma W and W electronic load. File, CHROMA Series, , Mainframes X, Operation Share technical documentation including product manuals, getting started guides, and.
|Country:||Republic of Macedonia|
|Published (Last):||12 February 2009|
|PDF File Size:||8.65 Mb|
|ePub File Size:||8.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
Chroma A Programmable Electronic Load integrates microprocessor capabilities into each load module and mainframe to provide simple and accurate parallel operation to optimize mnual speed and control among multiple load modules. Operation Table of Contents 1. Model 20 SPDT Relays Page 1 All technical data and specifications in this publication are subject to change without prior notice and do not represent a commitment on the part of Giga-tronics.
The Chroma Mainframe has room for four single-width Loads usee,or two double-width Loads Slew rate determines the rate at which maanual The RH measures and displays Temperature. In local mode any value can be set to a module from the keypad. The loading waveform is use in slew rates, load levels, duration and conducting voltage. Thank you for purchasing your new PSC More information. It is useful when a module is operating in CV or CR mode, or when it needs precise measurement.
Please refer to the System Installation Manual for information on limitations regarding product use and function and information on More information. Each terminal can carry 40 Amps at most. Official version will follow soon on: The low range provides better resolution at low voltage measurements.
Programmable DC Electronic Load. Documentation Brake module AX Please read this document carefully before installing and commissioning the brake module! When any protection occurs the module will turn off the load input, and produce beep sound till you remove the condition and reset protection by pressing LOAD key on the module. Updated versions of this document More information. There are two response speed of CV modes: Products You May Like.
Shall not be held liable for errors contained herein or direct, indirect. The rating cnroma on the device provides the class A or B of the equipment. Keep these More information.
Figure also illustrates a typical setup for remote sensing operation. The Chroma mainframe has room only for two single-width Loads or one double-width Load. Load modules occupy either one or two slots. Product Reviews Login or Register to write the first review.
It depends on the power rating of the module. The channel number is fixed for Mainframe even Load module is empty. Determine the position for chrpma Hands-free phone in the elevator phone. About this Manual We ve added this manual to the Agilent website in an effort to help you support your product. For example, maunal test of a battery charger can be simulated easily by setting the load to operate in constant voltage.
Click here to unsubscribe.
A/A Mainframe Rack Mount Kit: Chroma – Test Equipment Accessories and Cables
Power Electronics Testings Test Solution www. The programmable slew rate makes the simulation of transient load change demanded by real life applications possible. When local operation is in effect, you can select a channel, and use the display mankal well as keypad on the front panel to control the Load.
Start display at page:. One is latch, and another non-latch. Two or ussr load modules can be paralleled together to achieve the desired loading.
In remote mode programmed value cannot be chrooma boundary. All fields are required. The load will return to the previously programmed values when the Load is turned on again Protection Features Each load module includes the following protection features: The user can perform on line voltage measurement and adj u s tm en, ori lh c g used k ey p ad or ti n. Complain wrong Brand wrong Model non readable.
Easy programming More information.
Nothing contained herein shall. Chroma Manual Even though other OSes might be patible as well, we do not remend apying this release on atforms chroma a user manual other than the ones specified.
If input voltage uswr over the maximum of low range, you must select the high range.
Chroma 6314A 1200W Electronic Load Mainframe for 4 Load Modules
Rtp gold series manual rosin tech heat 5×5 press review Delta asme a 19 2 m toilet manual Evenflo triumph convertible car seat manual Fluid mechanics for chemical engineers second edition solution manual Haemonetics pcs2 user manual Nissan stagea manual conversion Foxconn n manual download Hp manual Ruger mini useg instruction manual Mechanical vibrations 5th edition solution manual pdf Audi a3 tdi owners manual Manipal manual of surgery 4th edition free download Pet fencing system manual Kenmore plus electric stove manual R statistics manual pdf Honda harmony en manual Godex label printer manual Perkins c 44tag2 parts manual Oracle user manual Kenwood kd owners manual Manual of section by paul lewis pdf Hp mini nr manual Introduction to electronics lab manual.
If any damage is found, please file a claim with the carrier immediately. Class B devices have a Federal Communications. In addition to this manual, be sure that the following items chrima also been received with your Mainframe and Load. Asus computer user manual Lcd cnroma 2 e manual Solution manual for fundamentals of microelectronics by behzad razavi victory vegas 8 ball service manual Canon zr80 manual pdf Honda hrv manual indonesia Kdxr service manual Canon cr service manual Severn glocon control valve manual Bissell prodry tlr power roller manual Rabbit service manual pdf Hsm 2 manual Allen bradley plc 5 programming manual Samsung manual Montgomery ward microwave oven manuals Audi a6 user manual Dell photo aio printer manual Dell openmanage manual chevy truck repair manual Casio salarms manual.
Programmable DC Electronic Load 6310 Series Operation & Programming Manual
When programmed data are over the boundary, Mainframe will set maximum or minimum level for the Load module. Overvoltage The overvoltage protection circuit is set at a level slightly above the voltage range specified in the specification of the Load. Therefore, both minimum transition time and slew rate must be considered in the determination of actual transition time.
If the CC mode of Load module is active, the new setting will immediately change the input at a rate determined by the slew rate setting.