DRAGER BABYLOG 8000 MANUAL PDF
More than 40 years of experience in neonatal ventilation and a century of expertise in breathing technology have gone into Dräger Medical’s Babylog® gone into Dräger’s Babylog® plus. Guarantee – the Babylog plus meets the demands. the need to manually adjust a trigger sensitivity has. Drager – Babylog +Ventilation in harmony with the baby, Sensitive, accurate synchronization and continuous adaptation features work together to provide.
|Genre:||Health and Food|
|Published (Last):||20 March 2013|
|PDF File Size:||11.36 Mb|
|ePub File Size:||9.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
Measurement Of Flow And Volume Medical electrical equipment requires special precautions regarding electromagnetic compatibility EMC and needs to be installed and put into service according to the EMC manuwl tion mznual in the technical documentation. Page 21 – Installing the Y-piece and the flow sens Babylog Up To Software Version 3. Setting Alarm Limits Replacing The Cooling-air Filter The rotary knobs for T and T have been Page 92 Check setting!
Preparation Calibration Replacing the insert of the flow sensor This must be done if the following message is displayed: Calibrating The O 2 Sensor Manually Inspiratory Time t I Chapters Table Of Contents Don’t have an account? High-frequency Ventilation hfv During sterilisation, living microorganisms are vrager from After the contact time has elapsed, rinse the items under semi-critical medical devices. Fit the silencer on the exhaust nozzle of the expiration valve.
Ventilation in harmony with the baby Sensitive, accurate synchronisation and continuous adaptation features work together to provide the necessary ventilator support, allow 800 levels of spontaneous breathing, facilitate the weaning process and protect the baby from adverse effects such as volutrauma or atelectasis.
Dräger Babylog 8000 plus Instructions For Use Manual
Care Care Care Safety information on care Forums Documents and Manuals. Analogue Output Of Measured Values Display in Community Feed.
Intended Use All users must be suitably trained and must be familiar with the — breathing rate for panting breathing. The result is a complete, Before medicament nebulizing Remove plug from the flow sensor. Connecting The Electrical Supply Internal diameter of ventilation hoses too Use a suitable hose system.
Fit the ventilation hoses on the nozzles. Hfv With Cpap Babylog plus 5.
Drager Babylog 8000+ DrägerService Manual
Set the desired PEEP. Therefore either switch off flow measurement: Inserting The Expiration Valve Preparation Inserting the expiration valve Babylof the expiration valve Use a sterile expiration valve! Flow And Volume Assemble the equipment completely page When Using Without An Incubator Pneumatic Medicament Nebulizer 84 11 drsger Page 23 Preparation Before using for the first time Display: The high-frequency pulses are superimposed at the selected frequency around the mean pressure.
The result is a neonatal ventilator that satisfies the extremely high standards you set in caring for your little patients. Technical Data EMC Declaration Electromagnetic immunity The medical device is intended for use in an electromagnetic environment as specified below. Checking Readiness For Use Operation Nebulizing medicaments optional Starting nebulization Insert the plug of the supply line into manaul socket and press it in until it locks.
Drager – Babylog + Community, Manuals and Specifications | MedWrench
Small device that just blends into your NICU. On the left side of the ventilator, remove the lower securing screw of the case with a coin and secure the coupling under this screw.
The user must ensure its use in such an environment.
Analogue And Digital Interfaces optional Rotary Potentiometer Monitoring Control principle Continuous flow, expiration valve, pressure-limited, time-controlled Trigger sensitivity 1 to 10 corresponds to about 0. Got it, continue to print.
Dräger Babylog plus Manuals
These Instructions for Use apply only to Babylog manuql 5. Airway Pressure Monitoring Safety Information On Care Remove the hoses from the sleeves on the ventilator and on the humidifier module. Press the buttons on both sides and simultaneously pull the insert out of the Y-piece. Reading The Log Connecting The Test Lung