ISO 10555-5 PDF
The main task of technical committees is to prepare International Standards. Draft International Standards adopted by the technical committees are circulated to. à aiguille interne. STANDARD. ISO. Second edition. Reference number. ISO (E). Provläsningsexemplar / Preview. STANDARD. ISO. Second edition. Reference number. ISO (E). This is a free 4 page sample. Access the full version online.
|Published (Last):||8 August 2013|
|PDF File Size:||8.24 Mb|
|ePub File Size:||3.85 Mb|
|Price:||Free* [*Free Regsitration Required]|
CSN EN ISO 10555-5
If a steel tube is used, it shall connply with ISO Designation and catalogue of CCR colours. Until that time, a manufacturer may label his product “radio- opaque” provided he can support this claim by demonstrat- ing that he has an appropriate method for showing radio- opacity.
Figure 1 — Typical over-needle peripheral intravascular catheter NOTE 2 At the time of publication of this part of ISOthere is no acceptable, validated test method to determine radio-detectability. I Principle Water is allowed to flow through the catheter and the amount of flow is measured either volumetrically or gravimetrically. The fluid pathway in the needle shall be free of unintended obstructions that 105555-5 prevent flashback.
The corresponding Indian Standards which 10555-55 to be substituted in their places are listed below along with their degree of equivalence for the editions indicated: This Indian Standard has been developed from Doc No.: Copyright BIS has the copyright of all its publications. I Principle A force is applied successively tensile and connpress- ive to the needle tube and needle hub and the tube- hub union is then examined for loosening. Certain conventions isk, however, not identical to those used in Indian Standards.
Bureau of Indian Standards BIS is a statutory institution established under the Bureau of Indian Standards Act, to promote harmonious development of the activities of standardization, marking and quality certification of goods and attending to connected matters in the country.
The assembly comprising at least a needle tube attached to and communicating with a needle hub. Annex D shows examples of typical needle point geom- etries.
EN ISO – European Standards
E-5 Test report The test report shall contain at least the following information: Fixed or is fitting per- mitting venting of air while restricting or preferably preventing the escape of blood. Suggested colour references for opaque naaterials are given in annex C.
This does not preclude the free use, in the course of implementing the standard, of necessary details, such as symbols and sizes, type or grade designations. NOTE To improve the visibility of the solution, a colorant such as red 105555- blue food dye may be incorporated.
Assembly comprising at least a needle tube attached to, and communicating with, a needle hub. Available from Munsell Color, N.
Fitting attached to iiso needle tube, providing communication with its bore. Other parts of this standard are as follows: In this adopted standard, reference appears to certain International Standards for which Indian Standards also exist. Users of Indian Standards should ascertain that they are in possession of the latest amendments or edition by referring to the latest issue of ‘BIS Catalogue’ and ‘Standards: Part 1 General requirements do The technical committee has reviewed the provisions of the following International Standard referred in this adopted standard and has decided that it is acceptable for use in conjunction with this standard: Rigid tube with one end sharpened to facilitate entry into body tissue.
Intravascular catheters ISO , -3, -4, -5
Review of Indian Standards Amendments are issued to standards as the need arises on the basis of comments. Measure the time taken for fluid to form the first falling drop at the back of the vent fitting.
An example of a suitable apparatus is shown in fig- ure B. Round the calculated average water fiowrate to the nearest whole number of millilitres. Page 3, subclause 4. Search the history of over billion web pages on the Internet.
Details of which is given in National Annex A. Blood flow into the needle hub. When the needle is fully inserted into the catheter unit, the catheter tube shall neither extend beyond the heel of the needle bevel nor be nnore than 1 nnnn fronn it see dinnension a in figure 105555-5.
Pages 1 and 2 4. Attention is particularly drawn to the following: