KEITHLEY 199 MANUAL PDF
Language English. Keithley: DMM Instruction Manual. Identifier keithley_DMM__Instruction_Manual. Identifier-ark ark://t8kd5tt7c. Model System DMM Scanner Instruction Manual Contains Operating and Servicing Information KEITH LEY WARRANTY Keithley Instruments, Inc. warrants . Keithley Pdf User Manuals. View online or download Keithley Instruction Manual.
|Published (Last):||19 September 2007|
|PDF File Size:||13.16 Mb|
|ePub File Size:||1.64 Mb|
|Price:||Free* [*Free Regsitration Required]|
When the selected trigger occurs, the storage process will commence. The integration period is dependent on usable 1999 as shown in Table Under these conditions, the FUR light blinks until the final, filtered reading is being displayed.
Keithley: DMM Instruction Manual : Free Download, Borrow, and Streaming : Internet Archive
Because of the rapid scan sequence in the scan mode, it is recommended that this mode be used with data store, as discussed in kfithley 2. Size The size of the data store can be controlled by one of the following I commands.
Describes the operation of the bus status indicators, and summarizes front panel messages that may oc- cur during bus operation. The average responding meter gives a TRMS reading of 5. This section contains information necessary to keithoey the instrument from the front panel.
Apply a voltage or manuak signal, that is to be used as the dB reference, to the input of the Model The display will zero and the ZERO indicator will be enabled. The following message is displayed briefly when a short time error occurs: The ZERO control can be used to null out constant off- set voltages.
Keithley: DMM 199 Instruction Manual
The Model will return to either factory default conditions or user saved default conditions. Select the ohms function by pressing the OHMS button. Translator Error Keuthley Often the user may incorrectly suspect that the hardware is causing a problem, when it was the software all along.
Two-Terminal Resistance Measurements Table If any errors occur, the instrument wiU display appropriate front panel error messages and generate an SRQ if programmed to do so.
OOdB on channel 1. The Model is then addressed to talk.
C Controller — The Model does not have controller capabilities. The following equation can be used to keiithley the voltage reference: Table lists the usable resolution on each function for the two S modes.
In a one-shot trigger mode, the Model will output a pulse once each time it completes a reading after be- ing triggered.
Quick disconnect screw terminals, 14 AWG maximum wire mnaual. To minimize problems in these areas, all input and output connections to the scanner card should be made using shielded cable when measuring low-Ievel signals.
Keithley 199 System DMM Scanner
Channels 2 through 8 referenced to Channel 1. The procedure below assumes that the default trigger soince has not been altered by programming the instrument over the IEEE bus. The frequency measured on the frequency counter is the high-end limit of the bandwidth. If the meter is connected to a high energy circuit when set to a current range, low resistance range or any other keithlet impedance range, the circuit is virtually shorted.
Full text of “Keithley: DMM Instruction Manual”
T6 X— Spaces are ignored. Cleared while the mstru- ment is processing commands. To modify the address, key in a new value with the numeric data buttons. Filter The Model uses two running average filters in order to reduce reading noise. Energize the circuit using the installed manua disconnect device and make measurements without disconnecting the DMM.
Note that the DCL command is not an addressed command, so all in- struments equipped to implement DCL will do so simultaneously. Place the DCL command byte on the data bus.